Brand Name : Mindray
Model Number : MEC-1000
MOQ : 1
Price : 1
Payment Terms : T/T, Western Union, Paypal
Delivery Time : 3-5 Working Days
Packaging Details : By Carton
Product name : Mindray
Type : MEC-1000
Warranty : 90days
Voltage : 110V—240v
Minimum purchase commitment : 1 unit
State : In Good Condition
Stock : In stock
Delivery Term : Express delivery
Language : English,French,Italian,Spanish,Japanese,Hindi,etc
Context of Use : ICU, emergency, Patient Monitor, ICU Room and operating room
Function : ECG,SpO2,NIBP,Medical Care,RESP
Packing : Carton
Lacation : Guangzhou, China
Mindray MEC-1000 Patient Monitor Power ON OFF Keypad Membrane
Mindray MEC-1000 MEC1000 Patient Monitor Power ON OFF Keypad Membrane Patient Monitor Parts
► Brief Introduction
Mindray MEC-1000 portable patient monitor has been designed to measure physiological parameters including ECG, RESP, TEMP, NIBP and SPO2, etc. Figure2-1 shows the structure of the whole monitor as well as the connection relationships between different parts. The board in the center of the figure is the core part of the monitor, i.e., integrated board for main control and parameter measurement, which, though being a single board, could realize the measurements of five said parameters; accordingly uniform AD conversion and digital processing system is used.
► How To Use
Locate the Power Button:Look for a button on the keypad or control panel that is labeled with a power symbol (often a circle with a line inside it) or with the words "Power" or "On/Off."
Press and Hold:In most cases, you may need to press and hold the power button for a few seconds to turn the device on or off. This is a safety measure to prevent accidental power cycling.
Follow Display or Indicator Lights:Pay attention to the device's display screen or any indicator lights. They often provide information about the status of the device, such as whether it is powering on or off.
Wait for Initialization:After turning on the device, it might take some time for it to initialize and be ready for use. Similarly, when turning it off, allow the device to complete its shutdown process before disconnecting or moving it.
Refer to the User Manual:For model-specific instructions, always refer to the user manual or technical documentation provided by Mindray for the MEC-1000. The manual will contain detailed information on the operation of the device, including the power on/off procedure.
Contact Mindray Support:If you encounter any difficulties or if the keypad functions are not clear, reach out to Mindray's customer support for assistance. They can provide specific guidance based on the model and version of the MEC-1000.
ORS beep and alarm sound
|Connected to central monitoring system
|10-level audible and visual alarm
Pls contact us, if you are interested in Mindray MEC-1000 MEC1000 Patient Monitor Power ON OFF Keypad Membrane .
Mindray MEC-1000 MEC1000 Patient Monitor Parts Patient Monitor Power ON OFF Keypad Membrane Images